Digi-Pas DWL-1500XY is an AFFORDABLE Wireless-Bluetooth, Dual-Axis Precision Digital Level of 0.001° resolution designed for entry level used by industrial users & professionals. DWL-1500XY replaces old-fashioned VIAL levels. Users read both X & Y axis levelling measurements wirelessly in proximity via Bluetooth, which effectively cut down installation & setup time effortlessly (one-man operation) for precision machines & equipment, especially when used in conjunction with "DigiPas Smart Level" Mobile App (Free one redeemable app with each purchase) with user’s smartphone (additional option to purchase multiple app available on Apple & Android App Stores).
Digi-Pas Precision Digital Level is highly accurate, verified by 3rd parties accredited calibration labs & test bodies in US, Japan & Germany in accordance with ISO/IEC 17025:2005 and ANSI/NCSL Z540-1 standard, traceable to NIST, JIS & DIN.
First, please download Smart Levels App from Apple or Google Store into your smartphone or tablet to enable FREE redemption of your app. If you who need more than one app to share a single device among multiple users, additional paid-apps can be purchased from Apple or Google Store.
NEXT, please follow steps below:
Tap "Redeem" on top right corner of App.
Gently scratch off the label on the back of the App Card accompanied to reveal the hidden 16-digit code.
Enter the 16-digit code & Company name then tap "Redeem" button.
"Redemption Successful" is prompted and you can use full features of the App now.
Switch angle display between X & Y Axis
Enable user to measure relative angles at a common plane with respect to a reference angle. The reference angle can be set using alternate zero feature
Enables user to perform absolute symmetry readings for both "front" & "reverse" (180 degrees) positions of any measurement angle.
Easy self-calibration feature allows user to reset the digital level back to factory preset accuracy.
Users may fully utilize Bluetooth mobile features for wireless levelling/alignment activities. One App is FREE to be redeemed for each purchase. Additional PAID Apps can be purchased from Apple and Google Stores. Each App works with the designated smartphone/tablet. Each App can connect with any device DWL1500XY (no restriction).
Precision levelling and angle measurement/alignment tasks essentially involve surface planes that demand for 2-dimensional solution rather than the outdated single-axis 'bubble' vial method that indicates 'bubble' leveling status 'one axis at a time'.
Why device screen only shown blinking "BT" indicator?
Blinking “BT” means the device has NOT yet been initialized. Users is required to initialize device using Digi-Pas Smart Level App downloaded either from Google Play Store or Apple App Store. Guide on how to pair your smartphone/tablet with device is Instruction Manual and also the App.
Why device reading is not zero when placing on flat surface?
This situation may happen when precision level is placed on a rough, dirty or uneven surface. Digi-Pas Precision digital level is very sensitive to pick up any surface unevenness. Ensure contact surfaces are clean and smooth to get consistent readings.
Why do I get two different readings when placing the device on front facing, and then replacing it on the reversed facing?
Your precision digital level may have been subjected to extreme physical or temperature shocks during usage, transportation or storage that affected its accuracy. An offset value of reading may have been present. Digi-Pas® precision digital level has a smart built-in feature that empower users to perform self-calibration to completely remove any residue offset and set the device to its factory accuracy. See video links on how to perform “User-Self Calibration”
Alternatively, users may just choose an easy ‘quick solution’ to simply perform "Absolute Level" Setting on the surface of the object to be measured to remove any residual offset value. Refer to Instruction Manual on HOW to perform "absolute level" feature.
Digi-Pas Smart Level App in my smartphone is unable to pair with the device. What should I do?
The device battery level might be low, please replace with new batteries.
Digi-Pas Smart level app unable to find Digi-Pas Device. What should I do?
Please ensure Digi-Pas Machinist Level Bluetooth is enabled to initiate wireless connection. To enable/disable device bluetooth on device、 Press and hold "Mode" button.
Ensure your Bluetooth Connection setting in your smartphone/tablet is also enabled.